Originally Posted by W3Rman
Is there any data I can find which show exactly how much faster the Rev.B is compared to the Rev.A? I may consider purchasing but only for significant improvement. I am sure someone must have both and could compare showing how AIO mode improvement.
Hi, according to feedback I have from a Ted's LightSpace CMS Calibration Disk user which is using LightSpace HTL + i1DisplayPRO Rev.B + KURO + eeColor and he has performed a lot of testing using LightSpace's 'Measure and log' where it exports in excel (CVS) file format the xyY and the measuring times per each read in milliseconds, for us to find by comparing the data's which mode and integration time settings is giving him better repeatability while it has faster measuring speed (Test Procedure Link
)....we found by comparing the data's between AIO vs. Burst Mode from i1DisplayPRO Rev.B Meter that that new AIO mode is about 30-50% faster from Burst Mode of Rev. A from the data I have.
The user tested 5% & 10% Luminance patches of Red/Green/Blue/Cyan/Magenta/Yellow/Gray and Black, which is the patches with the lowest luminance for him to use the 17-Point Cube DIP Mode Cube Chapter because darkest patches require more time to be measured from brighter ones and that test is required to help us decide what chapter to use for perfect sync of the disk with the software.
He used the 17-Point Cube DIP Mode Finder of Ted's Disk which displays for measurements the darkest patches that a 17-Point Cube (4.913) will measure, to be able to decide according to the measurements which DIP Mode setup he has to use...so with Rev. B meter he was able to run the 3 sec. Chapter of the 17-Point Cube Chapter of Ted's Disk where it performs 4.913 color reading and will take 4 Hours or 10-Point Cube (1.000 colors) that takes 50 minutes.
When I was using i1Display PRO Rev.A at past with my KURO using Burst mode, I was able to use the 6 sec. per color for the 17-Point Cube Chapter that was taking 8 Hours to complete the profiling. (Post-Calibration Report Link
So with Rev. B, it's real impressive the speed performance of the new AIO mode, which gives the capability of a Full Video Chain Profiling without using any external hardware or software (PC/Notebook) pattern generator.
If I will get feedback from other display technology user, I will inform you again.
Ted's LightSpace CMS Calibration Disk Free Version for Free Calibration Software: LightSpace DPS / CalMAN ColorChecker / HCFR
S/W: LightSpace CMS, SpaceMan ICC, SpaceMatch DCM, CalMAN 5, CalMAN RGB, ChromaPure, ControlCAL
V/P: eeColor 3D LUT Box -
P/G: DVDO AVLab TPG
Meters: JETI Specbos 1211, Klein K-10A, i1PRO2, i1PRO, SpectraCAL C6, i1D3, C5