Originally Posted by coldmachine
The machine has a CR spec of 2k min and 10k max. Thats 2k without DI and 10K with the DI. I believe the CR number is so low due to it being optimized for max light.
Wolfgang's CR numbers are within the normal margin of error for the specification. I know of another ownwr and his unit is around 9k. Personally I believe any manufacturer delivering 90% of the spec number is a very rare thing indeed.
my experience with christie is that are all numbers very conservative.
for sample if they spec. the hd6km with min. 2500:1 cr. normally
you will found least 2700:1.
thats not the case this time with this pr.and as you report now a other unit with the same number it seams that my test unit is not brocken.
ljg did you can see the same bug?
i drive it with this settings.
dynamic iris on.
to bad for me.
i had hope that this pr. can be the pr. i wait for
but i seams i have to wait ontill sim2 will have a pr. with the same
technology the lumis have but with mutch more light out.