Originally Posted by Chad B
What an improvement! Did you find that your first meter was defective?
Well I would have to assume so, I had repeatability issues with it (especially on red/magenta) where one day it was one reading and the next day it was different by a decent margin. I had been seeing erratic errors in CalMan and thought it was the software for about a week, now I am fairly convinced it was the meter. I was new to CalMan so having the questionable meter along with learning CalMan after using ColorFacts and HCFR for so long was 'interesting'.
I was able to slightly improve my Red down to dE 1.9 with a couple of ticks to match up luminance and just as before altering Red saturation would mess things up pretty good (on my 60LE632).
Once again thank you for your comments Chad, it most definitely encouraged me to seek a replacement meter. Now I just have to check it against my spectro (i1Pro). Oh and I do want to note that everyone at Spectracal has been AWESOME!