If anyone has any use for it, attached is a Pioneer KRP-500M matrix (thc file) for the D3 profiled against an i1Pro.
D3_i1Pro_PioneerKRP500M.zip 0k .zip file
I compared measurements using all the included and posted matrix/meter and .ccss correction files with my i1Pro. The results for the Pioneer when measuring grey scale with the D3 are as follows:
- the included "D3_generic_wide_ccfl_lcd" meter correction file was within .4 dE of my i1Pro, virtually identical. Surprising since it is an LCD correction file.
- the user-posted Panasonic Plasma 1 and 2 corrections were the only other matrix files that improved the D3 measures slightly, but were still off by about 4 dE.
- the rest of the meter correction files and all the ccss were less accurate then the D3 without any correction.
Great job on all the fixes and improvements Zoyd!