Originally Posted by guido310
So now i would like to use TED profiling, but i couldn't understand how to save the correction matrix after measurements.....any advice??
I cannot read initial meter setup instruction on displaycalibrations site.
Hello, maybe you mean how to use the Meter Profiling Check Layout Page that you found inside the 2 CalMAN 5.2 Workflows I send to my disk users.....
This layout page is for verification reason, it's useful for users to check if their meter profiling correction table is accurate.
I'm sure you will be surprised when you will find out that your familiar meter profiling procedure you followed since now maybe has create some not valid tables for some reason but you haven't realized that this is be possible because you have always followed the correct way but you haven't checked it.
To do this verification, start by using Reference Meter Page to take WRGB measurements using your Spectro and after that run your meter profiling procedure like you normally perform using CalMAN.
(re-measure again the WRGB patches using your spectro and colorimeter).
Later, use the Profiled Meter Page to measure the WRGB Patches with your Profiled Meter to see if the correction table is accurate vs. your reference meter readings.
You will found that sometimes it will need to re-perform your meter profiling 2-3 times to get you accepted results. Count also your meter repeatability/display instability here that maybe produce inaccurate meter correction tables.
About the Initial Page Setup Page of my site, you are right, It's not online yet but the instructions there will be for LightSpace Users, and I have posted these instructions at this post