A piece of electrical or painter's tape on the top of the bezel (holding the meter's cable in place) may work. Just be sure whatever tape you use is easily removable and doesn't leave any residue behind.
post #3091 of 3872
11/10/09 at 7:37am
Doug, it appears your posted .chc file is available here, right? And Color After 2 was your second pass?
Looking at the numeric data for the colors, you can see that your white and cyan Y measures are 124fL and 121fL, similar to your 100% white grayscale Y of 124fL. Yet the Yellow and Green Y values are 152fL. This is so off the mark that it screams out to me that you have some dynamic controls still enabled, probably CE dimming as you wrote that you have turned off dynamic contrast and energy saving. To my mind, this makes all the measures suspect.
Try 4 Midway 17.zip 0.8935546875k . file
pana_g15a.zip 3.5595703125k . file
panasonic_50pz800.zip 1.1240234375k . file
EllisSam8500aftercalday.pdf 138.6767578125k . file
Kino_Warm_Calibrated.zip 1.1103515625k . file