AVS Forum Special Member
Join Date: May 2004
Mentioned: 12 Post(s)
Tagged: 0 Thread(s)
Quoted: 736 Post(s)
As you can see, if I were doing the calibration run with black level compensation enabled, HCFR would have me raise Y, just as you suggested. The only difference being that HCFR is suggesting changes to all gray levels starting with large-ish amount at 10%, gradually decreasing to almost no change at 90%. The gamma chart changes as would be expected based on these re-calculated values.
220.127.116.11 is now up on sourceforge
+ Update Argyll code with fixes for i1d3 and i1d2 and support for Spyder 4
+ Add support for getting spectrum with Argyll meters
+ Add extra Argyll tools to install package
+ Add support for new display type function within Argyll
o Improve handling of multiple meters
o Fix issues with loading file with disabled meters
o Add missing driver files to installer
Could i1d3 and spyder 4 users try and use the new command line utilities to extract ccss files for their meters.
ccssFiles.zip 82.439453125k . file