Originally Posted by rachmat-d
Is it Possible to manually input the reference meter from jeti into calman enthusiast, for meter profile ?
Hi, you can do the following to create a meter correction table for CalMAN:
Open LightSpace DPS or HCFR or JETI LiVal (or ChromaPure...if you have ChromaPure Professional license level).....to operate JETI instrument from there, take WRGB patch readings, note the xyY readings per patch.
Open CalMAN read the same WRGB patches using your colorimeter and note the xyY readings per patch.
The pattern source has to be exact the same per each software, same RGB Triplet and same patch size. You can use my calibration disk for example where it has chapter for meter profiling:
For external patch generation users of CalMAN, they can open CalMAN and load the Ted's Meter Profile Check verification page:
...connecting their external generator and the colorimeter, and by just clicking each of the 4 patches from CalMAN Slider, CalMAN will send these patterns to generator, where the user can measure them with JETI using other software.
To the bottom left are of the workflow, there is a data0grid with the WRGB xyY data numbers, you can copy-paste that data quickly to CalMAN's Four Color Matrix Tool, using right mouse click etc.
Locate CalMAN Folder in your Start Menu and run the Four Color Matrix Tool utility:
Then copy-paste the WRGB xyY of JETI @ Reference Meter Data and WRGB xyY of Colorimeter you have @ Target Meter Data... the utility will generate a 3x3 XYZ matrix to enter these values manually from inside CalMAN later.
After having that 3x3 XYZ matrix data, use CalMAN, click to create a new meter profile:
Double click to any XYZ tab and copy-paste the 3x3 XYZ numbers that Four Color Matrix tool generated.
Doing all the above, you will have create a Four Color Matrix Correction table, operating the reference meter outside of CalMAN (when you don't have license level to operate a high-end spectro from inside your CalMAN).